This input module converts J, K Thermocouple and 4-20 mA Inputs to 0-2V Full Scale. These can be used for any voltage/current inputs too. The RTD module can be modified more easily for Voltage inputs. The control output can be On-Off or 4-20 mA/Proportional with another card. The 4-20mA I/O STC1000I is not complete in documentation.
This is a Input Signal Conditioning Card for the Temperature controller. The voltage levels from sensors are either too low or need to be translated in level and span. Then for greater accuracy some linearization methods have to be used for a more precise reading. This also increases the cost. The circuits here do no cover the linearization see others in this and my related pages.
The step or segment linearization can be done by transistor, diode or CMOS switches to accomplish varying attenuation/gain for stages of the curve or voltage levels. In Microcontroller systems it can be done by lookup tables or math.
In some older digital systems without a MCU, the A to D drives the address of an Eprom Array to get a Digital Data for Display, as a linearized Reading. This Corrected Data was in turn made into analog using a D/A and then on to a Chart Recorder. This was a Logic only System of the early days. Microprocessor systems was expensive, power consuming and use to frighten people by getting lost in loops or a short nap.(they have fixed that, make sure you code properly).
Input Module - J and K Thermocouple with 4-20 mA
PCB Boards of the Module